dental electrolytic polisher
Polishing unit dental lab equipment is used in restorative electrolytic polishing in dentistry. It can be used to polish objects that are polished by sandblaster such as cobalt-chrome alloys or objects made of stainless steel: denture frame, crowns and bridges and clasps. This polishing unit dental machine can also be used to polish work pieces that are made of special materials.
It has a delicate design and beautiful shape, its small size makes it a compact device. The dental electrolytic polisher baths are its cathode, and the anode is the clip which is used to hang casting. The timer controls the duration of electrolytic polishing, and the current can be controlled during the polishing process.
Voltage 220v/50HZ 10%
Output working current:0-20A can be adjusted
1.After unpack the package, check if all the accessories are inside. Read this electrolytic polishing in dentistry manual carefully before using.
2.Pour right amount of electrolyte into the baths.
3.Tie the casting to a metal wire and hang it on the L-shaped another bracket soak the casting completely into the electrolytic in the cathode ring. Do not let the casting on the anode touch the cathode to avoid short circuit. Then cover the organic glass.
4.Connect the machine to an electric outlet, turn on the power, then the power indicator should be on and the machine is in standby state.
5.turn on the power switch , adjust the timer clockwise to the desired time, turning the “time” button, then turning the current button, from small to large, start to polish, then there is foam in the slot.
6.After the temperature of liquid rises, it will causes bad smell, as this time, you should press the button “Heat radiation and smoke exhaust”, the pump and fan start to work.
7.After settimg the time, then it will automatically power off. At the same time, let the output current button to 0, take out the workpiece.
8. It is not allowed to get the workpiece without trun off the power during the process of polishing.
9.Because of careless operation, occur short circuit, then the fuse will be broke, you should turn off the electric to exchange the fuse.
10. The distance between the workpiece and polar plate should be more nearer and symmetrical; choose right liquid, setting the electrolytic time and current value, the temperature of liquid, the shape of workpiece, these parameter will decide the the polishing quanlity, so choosing carefully.
The working current should not be exceed 25A.
Sophie Michiels (verified owner) –
I also love that I can find all of our product listings on one website, instead of having to search for them on various different websites.
Gerald Phillips (verified owner) –
Thanks to Dental Laboratorio for providing a beautiful platform for dentist.
Osborne Bourdillon (verified owner) –
Alessandra de Aguiar (verified owner) –
Got it nicely and heavily packed.
Ананий Васильев (verified owner) –
The final product is beautiful and I am eager to get it in my office!